Determine the dynamic viscosity of sticky petroleum asphalts by vacuum capillary viscometer.It is also suitable to determine other materials which viscosity range is 0.0036～20.000Pa.s.
This instrument is made as per standards of Dynamic Viscosity Test for Bitumen (Vacuum Capillary Method), also meets SH/T0557-1998 Test method for Viscosity of Petroleum Asphalt (Capillary Method) .
It is used to determine the dynamic viscosity of sticky petroleum asphalts by vacuum capillary viscometer.It is also suitable to determine other materials which viscosity range is 0.0036～20.000Pa.s.
1.It has a data base for coefficients of capillary viscometers. It can save 10 groups of coefficients at most. Operator can callout the data whenever need.
2.It adopts high accuracy pressure sensor. The vacuum degree can be kept at 300mmHg±0.5mmHg. It can be shown on the screen clearly.
3.The system will ignore the results which timing is less than 60s and enter next measurement step. The system will calculate the viscosity automatically for the results which timing is beyond 60s.
4.Operator can choose and set automatic calculation of average value and bias percentage for 2~4 parallel tests at will.The calculation can be done for several times to remove the unsatisfied results for operator. The test report can be printed.
Temperature control mode
Point controlled by digital temperature controller.
Temperature control arrange
Temperature control accuracy
4 built-in timer. Can do timing for 4 capillary viscometers
0.0s～99999.9s (27.7h) Bias ≤0.02%
About 18 Pa.s～580000Pa.s
Touch-type colored LCD
AC (220±10%) V , 50Hz
32.5kg（no water in bath）
Maximum power consumption
Bitumen Dynamic Viscometer
Vacuum capillary viscometer type: 50,100,200
Mecury thermometer（50℃～100℃；scale mark 0.1℃）
Thermometer silica gel plug(1#)
Stopping ring of thermometer（1#）
Silicone tube（External diameter Ф12×thickness2）
Temperature control device usage manual